We ship globally and offer a range of shipping options, from drop-off delivery to white glove service including full installation. We work with a handful of shipping companies and have negotiated excellent bulk rates based on the large volume of freezers we move. So you can be assured whatever option you choose, you’re getting an […]
Shipping costs are determined by a handful of factors, including volume of purchase, shipping location and level of care. We can provide a firm, fixed quote prior to your purchase, so your final cost doesn’t come as a surprise.
Please contact Minus Forty Technologies toll free 1-888-725-3029 or visit us or try asking a Minus Forty Technologies Technician through our 24hr online support
Yes, a lock is available for each model. It is an optional item. For quotes and ordering, please contact our Sales Department at 1-800-800-5706, ext. 104; or 905-702-1441, ext. 104.
Troubleshooting guide is listed in Owner’s Manual for the freezer. Check following: Power cord unplugged or loose Faulty receptacle Circuit breaker tripped, or blown fuse, or open supply circuit Improper power supply applied to the freezer Freezer switch at the front turned off (13/22USG and 44UDG models only) Freezer in the 6-minute startup delay mode […]
Thermostat adjustments are explained in detail in Owner’s Manuals. On 13/22USG and 44UDG models, remove the front bottom grill to gain access to the electronic controller. Press P button on the controller, SP1 flashing will follow, then use arrow UP and/or arrow DOWN to increase/decrease the desired temperature. Press and release P to store the […]
Check following: Thermostat or electronic controller set too warm Freezer loaded with excessive amount of warm product Excessive door openings Insufficient clearance around the cabinet or freezer close to heating source(s) Improper power supply
Check following: Drain tube heater burned out or not sufficient Leak in the drain pan
Check following: Excessive ambient humidity Heated glass transformers not working or disconnected
Check following: Excessive ambient humidity and door openings The cabinet perimeter heater burned or disconnected Door and/or gasket not closing properly